R series of large rotary evaporator , mainly used in biology , medicine , chemical food and other fields of test , test and Production . Has a large capacity , large caliber rotary evaporation bottle , placed in a water bath , rotate the heating , so that the solution of diffusion efficient evaporation . Product can be used with circulating water type multi-purpose vacuum pump , diaphragm vacuum pump , low temperature cycle ( vacuum ) pump , circulating cooler , constant temperature cycle , low temperature coolant circulation pump and other ancillary components of the system
We have two kinds of conventional and explosion-proof type , can be select according to the experimental environmentTeflon ( PTFE ) and fluorine rubber double rotating seal , patent technology to ensure high vacuum
The temperature control system , accurate and reliable
In does not affect the system and solution under the condition of vacuum distillation . continuous collection
The PTFE discharge valve , corrosion resistance , no pollution
|Optimum ambient temperature℃||5~35|
|Working power supply(V/Hz)||Single phase 220/60|
|Main engine speed control||Digital frequency conversion stepless speed regulation|
|Mine engine speed(RPM)||10~140|
|Rotary electric machine(W)||40|
|Temperature control(℃)||Digital display temperature control,room temperature -99|
|Maximum vacuum(Pa)||399.9(Below 3mmHg)|
|Condenser tube||Vertical double coil|
The water is more than 2
The alcohol is more than 4
|Water bath pot material||Stainless steel|
|Water bath pot size(mm)||Φ300×170|
|Lifting function||Electric lift|
please connected properly ( see technical parameters ) , and ensure a good grounding .
work , cooling water pipe and exhaust pipe should be kept unobstructed , no dead corner .The glass component installation pipeline , should adopt push installation , so as not to damage the glass parts .
The use of flammable materials or organic solvents , please dont let the solution splash overflow , away from the fire and ready to take emergency measures . cleaning stubborn stains on the glass device , wipe clean with a rag , don't knock the lass with a hard object .
1.Rotary evaporator matching use with circulating water vacuum pump to offer a vacuum environment for reduced pressure distillation, they are widely used for vacuum filtration, vacuum evaporation, vacuum degassing, etc..
2.Rotary evaporator matching use with DLSB series low temperature cooling liquid circulating pump to conduct chemical reaction and medicine storage which requires low temperature environment.